Starting Loop: Settings
Settings are very important! Loop does math every 5 minutes to make decisions on raising basal, lowering basal, delivering a bolus, suspending delivery of insulin, and more. If your settings are not right, Loop will be making the wrong decisions.
Does this look familiar – it should. You’ll find these words repeated several times, in a variety of ways, throughout this website and the Diabetes Online Community.
This article provides a high-level answer to these questions:
- Why would my settings with Loop be different from my current settings?
- What if I’m switching to Pods from a tubed pump?
Why would my settings with Loop be different from my current settings?
Most people move to Loop assuming they can just transfer their settings from their pump or PDM.
This page goes over the three pillars: Basal, Insulin Sensitivity Factor (ISF), and Carb Ratio (CR).
There’s more to it than that, and you’ll find links to many settings suggestions as you keep going. Many of these suggestions are valuable whether you use a closed-loop system or pumping and and using a CGM in the “traditional” way.
- Loop uses the difference between your scheduled basal rate and what Loop has been giving you to calculate Insulin on Board (IOB)
- If your basal rate is too high
- You may have it set this way to cover meal portions (fat/protein) or small snacks
- Loop will think you need extra insulin when you don’t and your BG will tend to run below target
- If your basal rate is too low
- Loop will expect your BG to drop and may restrict basal
- Your BG will tend to run above target
- If your basal settings were really good pre-Loop, they will remain good
Insulin Sensitivity Factor
- Loop uses an insulin model, based on published results, for the type of insulin you select
- All insulin models have a long, low-level tail that stretches out ~6 hours depending on the type
- Manual pumpers only need to consider the first 3 to 4 hours of duration of insulin action (DIA) because they may be correcting at meals and maybe once between meals
- For the technically minded, the pump DIA model is a linear fall-off of activity, and does not include the exponential tail
- The small amount of insulin action in the long tail matters to the every-5-minute calcuations that Loop makes to predict future blood glucose out to DIA in the future
- Many Loopers discover they need to raise the number for their ISF (even if it was perfect when pumping manually)
- Loop allows you to input the expected absorption time for the carbs and when you plan to eat them
- Many loopers add in the effects of protein and fat along with carbs as an Effective Carb amount
- If you’ve been using Temp Basals or Extended Bolus to handle meals, you’ll need to learn a new protocol
- Once you figure it out – you will probably wonder how you ever did it before – but there is a learning curve
What if I’m switching to Pods from a tubed pump?
Many new Loopers switch to Eros Omnipod in order to Loop.
- Pro: you can use an in-warranty pump with Loop
- Con: if you have not previously been using pods, there may be a transition period
- You must use Eros pods
- You do not need to purchase a PDM
- Build Loop and use Loop to control the Pod
- Once you attach a Pod with Insulin to your body
- Use Loop in “Open Loop” mode to get your settings adjusted; this means that, at first, you will be pumping as any Omnipod user without the benefit of Looping while you get used to using the Omnipod system
- The cannula for the pod may provide different absorption at the site, so you may find that some of the settings you had with a tubed pump need fine tuning as you transition to to the Pods
- You might need to adjust your basal rate – get that done first; this might mean having to test your basal settings in detail
- Then work on your other settings
- Only when you are confident in your settings should you transition to Closed Loop
Please, take the time to read the docs, do testing in Open Loop mode and, once you switch to Closed Loop mode, pay attention.
Do NOT use the Walsh insulin model – it’s left over from the old days and will be removed soon. Select the insulin model for the type of insulin you are actually using.
The best references for getting prepared with your settings is this LoopDocs page.
LoopDocs will point you to LoopTips, where you should read and thoroughly understand these pages
Start to read these pages a little bit each day, as much as you can digest. You will keep coming back to these pages, and other pages and videos you find on this site.
Note that LoopTips is a wonderful resource written when an older version of Loop was in use. Be aware that the Loop you build may have different features, but the underlying information is still valuable.